User guide • Read online or download PDF • KYORITSU A User Manual • KYORITSU Tools. Get KYORITSU A User guide. Get all KYORITSU manuals!. Find great deals for Kyoritsu Digital Loop Tester Model A. Shop with confidence on eBay!.
|Genre:||Health and Food|
|Published (Last):||1 July 2007|
|PDF File Size:||9.27 Mb|
|ePub File Size:||12.36 Mb|
|Price:||Free* [*Free Regsitration Required]|
Tags Add Your Tags: No part of this material may be reproduced, translated, transmitted, framed or stored in a retrieval system for public or private use without the written permission of the publisher. In order to do it, can be used the RCD tester model A. Rating A 6 10 16 20 kyritsu 32 40 50 63 80 Disconnecting time 5s Disconnecting time 0.
This mode is cancelled whenever the test button is pressed. Get the item you ordered or get your money back. Featured Category Find more Fluke products,You can use side bar filter to look for your need easier and fast.
Tell us about it. Delivery times may vary, especially during peak periods. Kyoritsu A Loop Impendance Tester. Large custom digital display readout on all models. Your manual failed to upload If you find inaccurate product information, welcome to Correct Errors. All material on this site is subject to copyright. The RCD could trip during the tests also with model A.
The cause of the fault must be investigated and rectified. You have no items in your shopping cart.
Kyoritsu A loop impedance meter available from Test and Tag Supplies
It is fundamental for this example to test also the RCD to ensure that operation takes place quickly enough to respect the safety requirements. Page 13 In the event of a fault, the Fault loop impedance should be low jyoritsu and the Prospective Fault current higher enough in order to have the automatic disconnection of supply by the installed protection device within prescribed time interval.
Don’t see a manual you are looking for? Analogue Multimeters Digital Multimeters Accessories. This instruction manual contains warnings and safety rules which must be observed by the user to ensure safe operation of the instrument and retain it in safe condition.
KYORITSU 4120A User Manual
Such metalwork may become live for the duration of the test. Our Differences 1 Worldwide free express shipping, get your order free and fast. Add to Cart Qty: One of the significant features of this model is the D-Lok circuit that permits loop impedance testing without tripping some types of RCD installed in the circuit being loop tested.
The A also allows direct reading of the PSC prospective short circuit current.
A manufacturer developing, producing and selling electrical measuring equipment
Contact Us Order Status. Max value of Zs for this example is 2. Or point us to the URL where the manual is located. Be the first to kyorotsu this product Ask Question.
Page 14 The figure below shows in marked line the Fault loop impedance for TN system. Support Email support storeinfinity. It is able to measure low loop resistance to a resolution of 0.